Lick Counter DM 8Request InformationRequest Support
DM-8 monitors contact between animal and water bottle sipper tube. It is a classical implementation of electrical conductivity as a means of sensing contact. In this application, the animal stands upon a wire (stainless steel) floor in order to drink from a stainless steel sipper tube. An electrical pathway is completed that allows an imperceptible current (less than 1x10-6 ampere) to pass through the animal. Electronics within the DM-8 Control Unit sense the current flow, converting it into signals that are recorded using Columbus Instruments' Computer Counter line of interface equipment. In practice, each 'count' indicates one lick from the sipper tube. A DM-8 Control Unit supports up to eight drinking dispensers.
Drinking dispensers comprise a water bottle with sipper tube to which a wire has been affixed. The standard assembly has a volume of 300 mL and a sipper tube measuring 6 mm in diameter extending about 10 cm from the bottle stopper. Cages of various sizes are available with appropriately connected wire floors to provide a complete sensing assembly.
DM-8 and FM-4 are designed for compatibility and together provide a complete food intake monitoring system when configured with the appropriate Columbus Instruments Computer Counter Interface.
Features / Specifications
Dispenser Volume: Approximately 150ccm
Sensing Current: less than 1 microampere
1030 DM-8 Drinking/Lick Monitor Control Unit for 8 Dispensers
1035R DM-8 Drinking/Lick Monitor Cage for Rat
1035M DM-8 Drinking/Lick Monitor Cage for Mouse
1040 DM-8 Drinking/Lick Monitor Water Dispenser
1060 DM-8 Drinking/Lick Monitor Economical Computer Interface for 8 Dispensers (GPCmB-8)
Click on any of the following titles to search for the document in a new window using Google Scholar.
Circadian Rhythms and Patterns of Running-Wheel Activity, Feeding and Drinking Behaviors of Old Male Rats
Some of these references may be difficult to locate using Google Scholar. If you are having trouble finding the correct reference paper, try searching for it on PubMed: http://www.ncbi.nlm.nih.gov/PubMed
Peng M.T. and Kang M.
Physiol. & Behav. Vol.33(4):615-620 (1984).
Manipulation of potential perinatal zeitgebers for the juvenile circadian temperature rhythm in rats
B. Nuesslein-Hildesheim and I. Schmidt
Am J Physiol Regulatory Integrative Comp Physiol, Nov 1996; 271: R1388 - 1395.
Reward without dopamine.
Cannon CM, Palmiter RD.
J Neurosci. 2003 Nov 26;23(34):10827-31.
Establishing Meal Patterns by Lickometry in the Marmoset Monkey (Callithrix jacchus): Translational Applications From the Bench to the Field and the Clinic
CORINNA N. ROSS, MICHAEL L. POWER, SUZETTE D. TARDIF
American Journal of Primatology
Volume 74, Issue 10, pages 901-914, October 2012
Cre recombinase-mediated restoration of nigrostriatal dopamine in dopamine-deficient mice reverses hypophagia and bradykinesia
Thomas S. Hnasko*, Francisco A. Perez*, Alex D. Scouras, Elizabeth A. Stoll*, Samuel D. Gale*, Serge Luquet, Paul E. M. Phillips, Eric J. KremerÂ¶
, and Richard D. Palmiter,||
PNAS | June 6, 2006 | vol. 103 | no. 23 | 8858-8863
Salt consumption increases blood pressure and abolishes the light/dark rhythm in angiotensin AT1a receptor deficient mice
Yanfang Chen, Terry L. Oroszi and Mariana Morris
Physiology & Behavior Volume 88, Issues 1-2, 15 June 2006, Pages 95-100
Animal models of depression in dopamine, serotonin, and norepinephrine transporter knockout mice: prominent effects of dopamine transporter deletions
Perona, Maria T.G.a
; Waters, Shonnaa
; Hall, Frank Scotta
; Sora, Ichirob
; Lesch, Klaus-Peterc
; Murphy, Dennis L.d
; Caron, Marce
; Uhl, George R.a
Behavioural Pharmacology:Volume 19(5-6)September 2008pp 566-574
Starvation after AgRP neuron ablation is independent of melanocortin signaling
Qi Wu*, Maureen P. Howell, Michael A. Cowley, and Richard D. Palmiter*
PNAS February 19, 2008 vol. 105 no. 7 2687-2692